Section 7. PROPER CARE AND
MAINTENANCE FOR THE PARI DURA-NEB 3000
THE DURA-NEB 3000
If you plan on storing the DURA-NEB
3000 for an extended period of time, recharge the internal battery overnight once every 2
months, unplug the unit, then return the unit to storage. Under normal circumstances, as
the battery nears the end of its life, it will require recharging more frequently. Contact
your DURA-NEB 3000 dealer (AMERICAN ALLERGY SUPPLY 1-800-321-1096 )or PARI Customer
Service when you notice diminished battery capacity. Servicing is strongly recommended
when the battery is not operating properly.
Note: Excluding battery, there are no
user serviceable parts inside the DURA-NEB 3000 compressor case. Service by a
non-authorized DURA-NEB 3000 service center voids warranty.
AIR FILTER REPLACEMENT
We Recommend Replacing the Filter
Every 6 Months with Daily Use. Price $15 for 5 filters
Pari Part # 41F64 .
Under normal conditions, the
filter on the front panel of the DURA-NEB 3000 should be replaced after 6 months of use,
or sooner if it appears soiled (grey), clogged or wet. Check the condition of the filter
after every 10-12 treatments. Do not attempt to wash and re-use filters. Do not use
substitute filter materials. Compressor damage can result. Any attempt to use alternative
filter materials will void compressor warranty and cause damage to the compressor. When
replacing the filter, make sure that the front panel on/off switch on the DURA-NEB 3000 is
in the "OFF" position. Turn the round plastic filter cover counterclockwise
(left) and remove filter (use a coin to help turn the cover if tight). The entire filter
assembly was designed as a remove and replace item (Fig.E-12). That means removing the
filter, discarding it, and replacing it with a new filter assembly (part # 41F64 ). Note: Do not attempt to disassemble filter
assembly. Do not attempt to wash and re-use filter material. Do not attempt to replace the
filter material only.
Section 8. PROPER CARE AND MAINTENANCE
CARE AND MAINTENANCE
CLEANING THE DURA-NEB 3000 HOUSING
Use a damp cloth to wipe
the exterior of the DURA-NEB 3000 housing. DO NOT USE ANY COMMERCIAL CLEANERS ON THE UNIT.
Never operate the
DURA-NEB 3000 if it is damaged or improperly or incompletely assembled. Use only original
PARI parts and attachments authorized by PARI for the DURA-NEB 3000. Use of other parts
and attachments may damage the unit or cause improper treatment. Do not open or
disassemble the compressor unit (except for the Battery Access Door) at any time. Repairs,
service and maintenance should be carried out only by authorized PARI service centers.
CLEANING THE TUBING
If there is any moisture
or condensation in the tubing, let the compressor run with only the tubing attached for 10
minutes. The warm air from the compressor will dry the moisture or condensation.
Alternatively, you can dry the tubing by removing it from the compressor and hanging it up
with the "ends down" (Fig.E-11) to allow any moisture or condensation to drain
out and evaporate. Use a damp cloth to wipe the exterior of the tubing.